These are chat archives for nightscout/intend-to-bolus
rm -rf /root/src/oref0/to remove it, and then
cd /root/src/ && git clone https://github.com/openaps/oref0.gitto re-clone
myopenaps/autotune/autotune_recommendations.logthat’s the easiest way to see what the basal rates are. The other way would be to look in
If it says 1.222, it’s going to use that number as your baseline basal for 10am for all the calculations about how much more or less insulin you need for the time frame. It’s not a daily dose report; it’s a tuned profile that’s then used until the next run when it tunes further.
yes, the autosens ratio is applied to your autotune profile
the APS - system is perfect for my sister. But now, before we start to build a rig, i got a question to the group.
She use the Medtronic Enlite Glucose CGM's with a Medtronic minimed Paradigm 712 insulin pump.
Is it okay for the rig - system or could we get some trouble with this combination ?
I think, it will be okay.
Thanks a lot..
Hi all :)
I am really impressed with 0.7.0 - thank you for everyone's effort.
I am finding that it is a little aggresive with its insulin amount. For example, over 3 - 4 hours my blood will go from 8.9 down to 3.3 and my pump for example right now is giving 1.9 units per hour as a basal, and my blood is 5.9. To me that is too high, it should be a lot less, because if it continues to give me that, im definitely going to be taken low.
What shall I adjust to fix this? I can go ahead and change my base basal rates...is that the best idea?
Thanks a lot for the information.
For a newbie is it not so easy to find the right combination.
But in our case, we prefer an offline system, if it is possible.
My sister have two pumps. The first one is the 712. And the second one is really a 640G.
But...for a test, we could take the hotspot funktion of the smartphone to send the data to the cloud.
For my own appreciation: The CGM sends the BG to the pump and then, in our case, the RPi Zero with Adafruit-Shield load it from the pump ?
Sorry for these unusual question, but i try to understand the system.
Thanks a lot
So, if we want to use the 640G pump with the Enlite's, what is a CNL meter ?
Thank you...you make our day !!