@juanjuanmejias_twitter hm, have you looked at changing the display in NS? (I don’t know if it’s possible but I’m guessing NS would be the easiest place to change?)
@acember no (isf gets adjusted in bolus wizard settings, then picked up in next autotune run) unless you have some of the exercise setting modes enabled, which means a target adjusts your sens ratio manually which then adjusts isf as well.
Read the news release yesterday on the new FDA classification, ACE, allowing for interoperability with the Tandem pump. And making it now possible to communicate and receive dosing commands from other devices or AID's. Problem is no devices are listed or approved. Seems like this is step 1 to get to a faster authorization for the Tslim to close the loop by itself or with add-on devices. ... like our Rig. https://www.fda.gov/NewsEvents/Newsroom/PressAnnouncements/ucm631412.htm
It’s unlikely they’ll make it possible to use with a non-regulated controller (eg rig). So yes, nice big step toward interoperable commercial devices & future choices. Little to zero impact expected for DIY, though
@ecc1 Maybe I'm being a spaz (and/or just not used to reading JS), but can you point out the lines that are doing the math here? Lines 105/106 or 110/111 which update the variable values look like they just come out of nowhere. Is detect(detection_inputs) the function that actually does something?